Many of the world’s 50,000 foundries rely on optical emission spectrometry (OES) to give them accurate and reliable one-step quality control throughout the melting process. By providing a quick and simple way to see the exact chemical composition, instruments like the Hitachi High-Tech range of OES analysers increase productivity and efficiency.
Keeping your spark spectrometer performing at a high level means keeping an eye out for the early signs that some maintenance might be necessary.
One of the biggest indicators that your OES is not performing at the level you need it to be is the appearance of bad burn spots.
A bad burn spot is evidence that the discharge is not concentrated. A diffuse charge means that your analysis is likely to be of little use to you.
There are two common types of bad burn spots that you need to look out for. If the centre of the burn has not re-melted or if the edge appears milky white, you have a bad burn on your hands and your OES needs some maintenance. It is a sign of a dip in your spark spectrometer’s performance.
Unlike the results of a diffuse discharge, a good burn spot has a clearly re-melted centre and a strong black circle of deposit around it.
Put simply, bad burn spots mean that your spark spectrometer is not working in the way that it should be. This means that the analysis it is returning is unlikely to be accurate. For foundries that rely on keeping a close eye on the exact chemical composition of the products they are making, faulty analysis will quickly become costly.
If you are getting bad burn spots, there are a couple of things that you should check first.
Your first port of call should be to take a look at the materials you are analysing. Specifically, if there are any inclusions, blow holes or cracks.
The problem could be in the way you are preparing your samples. Accurate OES analysis requires good and careful sample preparation.
Another common cause of bad burn marks is the quality of the argon you are using or the way it is being supplied to your OES analyser. 4.8 grade Argon is the required minimum which should be used with an OES. 5.0 grade argon or higher is the grade that should be used for spark spectrometry when you need to determine gases such as Nitrogen within your material as it is very unlikely that other contaminates will be present. The amount of pressure within an argon bottle is also important to ensuring premium performance. Please refer to you argon supplier’s recommendations regarding the minimum pressure you should reach before replacing your bottle.
The tubing of the argon delivery system could also be to blame. Any leaks could be allowing in outside gases to contaminate the system. Obstructions in the tubing could also be stopping debris from the sample area from being cleared away properly.
Once you have checked your spark spectrometer’s set up, you should perform a number of burn samples to make sure that your analysis is back to the level you need it to be. We would recommend that you do at least three to make sure.
With over 45 years of experience in providing cutting-edge materials analysis to companies across the world, Hitachi High-tech knows all about getting the best performance out of analysers. We provide a range of OES analysers, meaning that we have a solution to match your operational needs. We support our clients throughout the life cycle of the solutions we provide so that you can always be sure that our analysers are adding to your operations rather than detracting from them.
To get more information on the Hitachi High-Tech range of OES analysers, or to book a demo, contact a member of our team today.Find out more
XRF Analysis for Better Coil Coating Quality ControlRead More
What makes the FT110A the best analyser for your coatings measurements?Read More
Did you know this year’s 2019 U.S. tax deduction has many advantages for small to medium businesses investing in growth? Here’s what you need to know.Read More